Bovine Hemophilia Factor Antigen (vWF-Ag) Quantitative Detection Kit (ELISA) Instruction Manual [Kit Name] Bovine Vascular Hemophilia Factor Antigen (vWF-Ag) Quantitative Detection Kit (ELISA) [Reagent Kit Usesã€‘ Quantitatively detect the content of von Willebrand factor antigen (vWF-Ag) in bovine serum, plasma and related liquid samples. [Detection principle] This kit adopts double antibody one-step sandwich enzyme-linked immunosorbent assay (ELISA). Add the standard, the sample to be tested and the enzyme-labeled working solution to the transparent enzyme-labeled plate precoated with bovine hemophilia factor antigen (vWF-Ag) antibody. After incubating for a sufficient time, wash to remove unbound Ingredients. Add the developer A and B in sequence. The developer (TMB) is converted into a blue product under the catalysis of horseradish peroxidase (HRP), which turns yellow under the action of acid. The concentration of von Willebrand factor antigen (vWF-Ag) was positively correlated. The OD value was measured at a wavelength of 450 nm. Based on the OD value of the standard and the sample, the bovine von Willebrand factor antigen (vWF-Ag) content in the sample was calculated. [Composition of the kit] 1 Enzyme label coated plate 12 wells Ã— 8 strips 7 Developer A solution 6mL 2 Standard product: 240U / L 0.6mL 8 Developer B solution 6mL 3 20-fold concentrated washing solution 25mL 9 Stop solution 6mL 4 Standard product Diluting solution 6mL 10 Instructions 1 copy 5 Sample diluting solution 6mL 11 Sealing film 2 sheets 6 Enzyme labeling reagent 10mL 12 Sealed bag 1 Remarks: The standard diluent used for standard products is sequentially diluted as: 240, 120, 60, 30, 15 , 7.5 U / L [reagents and equipment not needed] 1, 37 â„ƒ incubator 2, standard microplate reader 3, precision pipettes and disposable tips 4, distilled water 5, disposable test tubes 6, water absorption Paper [Operation Steps] 1. Preparation: Remove the reagent kit from the refrigerator and re-equilibrate at room temperature for 30 minutes. 2. Mixing solution: dilute the 20-fold concentrated washing solution with distilled water to the original one. 3. Add standard products and samples to be tested: take a sufficient number of enzyme-coated plates and fix them to the frame. Set up standard wells, sample wells to be tested and blank control wells, record the positions of each well in the standard wells. Add 50Î¼L of the standard; add 10Î¼L of the sample to be tested to the sample well, and then add 40Î¼L of the sample diluent (that is, the sample is diluted 5 times); add 100Î¼L of enzyme-labeled working solution to each well; no blank control well. 4. Incubation: Incubate in a 37 Â° C water bath or thermostat for 60 minutes. 5. Wash the plate: discard the liquid, pat dry on the absorbent paper, fill each well with the washing liquid, let stand for 1min, shake off the washing liquid, pat dry on the absorbent paper, repeat the washing of the plate 5 times (you can also use the washing machine to press Instructions for washing the board). 6. Color development: add 50 Î¼L of developer A solution to each well, then add 50 Î¼L of developer B solution, and develop color at 37 Â° C in the dark for 15 min. 7. Termination: Remove the enzyme labeling plate and add 50Î¼L of stop solution to each well to stop the reaction (the color changes from blue to yellow). 8. Determination: Zero the blank holes, and within 15 minutes after termination, measure the absorbance (OD value) of each well with a wavelength of 450 nm. 9. Calculation: According to the concentration of the standard product and the corresponding OD value, calculate the linear regression equation of the standard curve, and then calculate the corresponding sample concentration on the regression equation according to the OD value of the sample. You can also use various application software to Calculation. The final concentration is the actual measured concentration times the dilution factor. [Sample requirements] 1. The sample must not contain sodium azide (NaN3), because sodium azide (NaN3) is an inhibitor of horseradish peroxidase (HRP). 2. The specimen should be extracted as soon as possible after collection. The extraction should be carried out according to relevant literature. The experiment should be carried out as soon as possible after extraction. If the test cannot be performed immediately, the specimen can be stored at -20 â„ƒ, but repeated freezing and thawing should be avoided. 3. The sample should be fully centrifuged, without hemolysis and particles. [Notes] 1. The experiment is carried out in strict accordance with the instructions, and the result of the experiment must be determined by the reading of the microplate reader. 2. If the enzyme-labeled coated board is not used up after opening, it should be put into a sealed bag and added with desiccant immediately. 3. It is recommended that all standards, samples and blank controls be tested in duplicate, and the average value is taken to reduce the experimental error. 4. If the color is too light, the substrate incubation time can be extended properly. 5. In order to avoid cross-contamination, the standard, sample and blank control should be replaced with a tip for each additional one; the common components such as enzyme working solution, sample diluent and substrate should be added by cantilever, and should not touch the microwell ; Do not reuse the sealing film. 6. The kits are used within the warranty period, and different batches of reagents should not be mixed. 7. Substrate B is sensitive to light and avoid prolonged exposure to light. [Summary of operating procedures] Prepare reagents, samples and standards, add the prepared samples, standards and enzyme standard working solution, wash the plate 5 times at 37 Â° C for 60 minutes, add color developing solutions A and B, and develop at 37 Â° C for 15 minutes Calculate the OD value within 15 minutes after adding the stop solution. [Detection range] 7.5-240 U / L [Specification] 96 servings / box [Storage] 2-8 Â° C, protected from light and moisture. [Validity period] 6 months Shanghai Yuping Biological Technology Co., Ltd. mainly deals with ELISA kits of various brands and grades, with quality assurance and perfect after-sales service. And provide free generation testing. Serving universities and immunology research units. Technicians serve you better.
Indoor Playground is booming business, As middle class population growing fast, Parents do not have much time on work time, more than that, all parents do best to make they children have a wonderful time after school, the demand of children service keeping grow. That bring huge opportunities for indoor playground business.
Golden Childhood Playground Equipment CO.,LTD bring you the most fashion design like Ice worlds, Space ship, Castle and large combination play centre.
Golden Childhood playground equipment CO.,LTD is one of the top level China indoor playground & manufacturers and suppliers, we are always able to provide customers with high quality indoor play equipment and soft play equipment.
What you need to prepare before have a good design for your own business? .
Commercial Indoor Playground Equipment,Indoor Playground,Large Indoor Play Structures,Theme Playground Equipment
Shenyang Golden Childhood playground equipment CO.,LTD , https://www.outdoor-playground.com